|Package:||Bottle Or Cubitainer In Carton|
human biochemistry reagents,
chemically pure reagents
Cell Counter Reagents for LANDWIND 3 Part Hematology Analyzer LW-D3600 Compatible Diluent Lyse Rinse
Intended Use To be used for blood cell counting, sizing, and WBC differentiation, hemoglobin determination on LANDWIND LW-D3600 Hematology Analyzers.
|Cat. No||Description||Pack Size|
This product is for In Vitro Diagnostic Use only.
Re-calibration is recommended when changing the origin of the reagents.
It is not recommended using different brands of solutions together on one unit of analyzer.
Do not ingest. Avoid contact with skin and eyes.
Skin contact: Flush with plenty of water.
Eyes contact: Flush with plenty of water. Obtain immediate medical attention.
This kit is for professionals and in vitro diagnostic use only.
Never put distilled water on the analyzer (except clean for shipping menu), otherwise the pneumatic unit can be damaged seriously.
Will the quality of product change with different temperature?
Normal weather change will not affect the use of the product. If the reagents freeze for low temperature, returning it to room temperature make the ice melt completely and the products can be used normally. Ensurind the ice melt completely otherwise it may cause blockage. The store temperature of hematology reagents is 2-35 degrees, and the biochemical reagents require 2-8 degrees, if not the validity of products will shorten.
Can reagents cross use with different manufactures?
Some reagents can cross use but some need to use as a set otherwise the results may be effected. For example, LYSE-4DL, LYSE-4DS,NR-PACK and RET-PACK of SYSMEX XE-2100 model should be used as a set.