|Type:||Diluent Lyse Dye||Sample:||Blood|
|Expiry:||24 Months||Package:||Cubitainer And Carton|
chemistry analyzer reagents,
human biochemistry reagents
Hematology Analyzer Reagent SWELAB AC970 CBC 3 Part Diluent Rinse Lyse Cleaner
Specification of Swelab Reagents
|Compatible Reagents on Swelab AC970 AC920 AC910 AC900|
|Cat. No||Description||Pack Size|
Intended Use of Hematology Reagents To be used for blood cell counting, sizing, and WBC differentiation, hemoglobin determination on SWELAB AC970 Hematology Analyzers.
Close system: The manufacturers impose restrictions on software by scanning their barcode when changing reagents to prevent other compatible reagents.
Open system: The manufacturers do not impose restrictions on software so that reagents can be used no matter what brands.
Storage of Swelab CBC Reagents
Store the product at 2-35℃ in a dust free place with good ventilation. Temperature outside the range may damage the product.
The open vial life of the product is 60 days. Unused reagents should be discarded after 60 days. Do not mix residual with a new carton.
Should the Blood Cell Counter instrument do a calibration before using our reagents?
First of all,if your instrument used the original reagents before using our products, you needn’t to do calibrate.
If your instrument used other compatible reagents, calibration is necessary before using our reagents.
Limitations of Detection Method
If the temperature of the reagent is out of the specified range, the specimen's measurement parameters may appear abnormal. If in this condition, please confirm the measurement parameters with the microscopy method. Please refer to the instrument operation manual for details.
The product should be stored in 2℃-35℃ and the storage period is 2 years. After opening the bottle, it is used in 15℃-30℃, and the validity period is 60 days.