|Place of Origin:||China|
|Minimum Order Quantity:||500 units|
|Packaging Details:||20L, 5L, 500ml, 1L, 50ml, 42ml|
|Delivery Time:||in 20 days|
|Payment Terms:||L/C, T/T, Western Union|
|Model:||JASDAQ DH-510||Shelf-life:||24 Months|
|Usage:||Medical Use||Differential:||3 Part|
human biochemistry reagents,
chemically pure reagents
Cell Counter Chemically Reagents For JASDAQ DH-510 DH-520 DH-560 Hematology Analyzer with ISO/CE/CFDA approvel
Intended Use To be used for blood cell counting, sizing, and WBC differentiation, hemoglobin determination on JASDAQ DH-510 DH-520 DH-560 Hematology Analyzers.
|JASDAQ DH-510 DH-520 DH-560|
|Cat. No||Description||Pack Size|
Re-calibration is recommended when changing the origin of the reagents.
It is not recommended using different brands of solutions together on one unit of analyzer.
Do not ingest. Avoid contact with skin and eyes.
Skin contact: Flush with plenty of water.
Eyes contact: Flush with plenty of water. Obtain immediate medical attention.
This kit is for professionals and in vitro diagnostic use only.
Never put distilled water on the analyzer (except clean for shipping menu), otherwise the pneumatic unit can be damaged seriously.
Store the product at 2-35℃ in a dust free place with good ventilation. Temperature outside the range may damage the product.
The open vial life of the product is 60 days. Unused reagents should be discarded after 60 days. Do not mix residual with a new carton.
3. Is there great difference of types between different kinds of hematology reagents?
Different analayzers have different using principles, therefore there is a great difference of types and quantities. In general, the types and quantities of reagents for use of five part hematollogy analayzers will more than three part machine.