|Sample:||Blood||Usage:||Medical, Clinical Experiment|
chemistry analyzer reagents,
human biochemistry reagents
Hematology Analyzer Reagent for Abbott CD1800 CD1700 CD1600 CD1400 CD1300 3 Part
Intended Use To be used for blood cell counting, sizing, and WBC differentiation, hemoglobin determination on ABBOTT CD-1800 CD1700 CD1600 CD1400 CD1300 Hematology Analyzers.
This product is for In Vitro Diagnostic Use only.
Frequently Asked Question
Which Lyse can test EOS and BASO?
Firstly, Three Part hematology analayzers can’t recognise EOS and BASO.These kind of machines are currently used in one detecting principle--- electric impedance method. Although the brand of analayzers are different, their reagent species are broadly similar. Reagents can be classified to a kind of Diluent, lyse and two to three kind of washing solutions. Only Five Part hematology analayzers can recognise five differential of leukocyte, which are monocytes, lymphocytes, neutrophils, basophils and eosinophils. Different brands of five part analayzers are different in principles and their effects and types of hemolytic agents are also different.
|Abbott||CD3500/3700/3000||WIC/HGB LYSE||WIC/HGB LYSE|
|Abbott||CD3200||WBC LYSE||WBC LYSE|
|electric impedance method test BASO, no exclusive lyse|
|COULTER||GEN.S, STKS, MAXM, HmX||LYSE-DIFF|
|COULTER||AcT 5diff, AcT 5diff AL, AcT 5diff OV||
|5 LYSE-DIFF HGB(electric impedance method test BASO)|
Re-calibration is recommended when changing the origin of the reagents.
It is not recommended using different brands of solutions together on one unit of analyzer.
Do not ingest. Avoid contact with skin and eyes.
Skin contact: Flush with plenty of water.
Eyes contact: Flush with plenty of water. Obtain immediate medical attention.
This kit is for professionals and in vitro diagnostic use only.
Never put distilled water on the analyzer (except clean for shipping menu), otherwise the pneumatic unit can be damaged seriously.
Store the product at 2-35℃ in a dust free place with good ventilation. Temperature outside the range may damage the product.
The open vial life of the product is 60 days. Unused reagents should be discarded after 60 days. Do not mix residual with a new carton.
|Abbott CD1800 CD1700 CD1600 CD1400 CD1300|
|Cat. No||Description||Pack Size|