|Store:||Room Temperature||Package:||Cubitainer And Botte In Carton|
mindray hematology analyzer reagents,
chemistry analyzer reagents
Laboratory Mindray Hematology Reagents 3 Part BC-2300 BC-2100 BC-2000 Compatible
Medical Use of Mindray Reagents
To be used for blood cell counting, sizing, and WBC differentiation, hemoglobin determination on Mindray BC-2300 BC-2100 BC-2000 hematology analyzers. This product is for In Vitro Diagnostic use only.
|CBC Reagents on Mindray BC-2300 BC-2100 BC-2000|
|Cat. No||Description||Pack Size|
Pay Attention of CBC Reagents
1. Pay attention to seal and prevent pollution;
2. It is strictly forbidden to eat and avoid contact with eyes and skin. Once Contact your eyes and skin,please clean immediately with plenty of water and seek medical advice.
3. Read the instructions carefully before use, and should stop using them when they exceed the period of validity;
4. If the product freezes, it should be completely defrosted at room temperature and used after mixing.
5. Waste liquid, waste, residual products and treatment of contaminated packaging materials, please comply with local regulations;
6. This product is an in vitro diagnostic reagent;
Limitations of Detection Method
If the temperature of the reagent is out of the specified range, the specimen's measurement parameters may appear abnormal. If in this condition, please confirm the measurement parameters with the microscopy method. Please refer to the instrument operation manual for details.
Shelf Life and Storage of Cell Counter Reagents
The product should be stored in 2℃-35℃ and the storage period is 24 months.
After opening the bottle, it is used in 15℃-30℃, and the validity period is 60 days.
What is the difference of Rinse and Cleaner?
The washing solutions can be classified to two kinds. One is to clean the pipeline after a single test to prevent cross-contamination, and to use after each test; Another is generally alkaline (such as the Mindray Probe Cleanser), which is used to shut down (for example, Mindray's e-z Cleanser) and regular maintenance.
Closed system: The manufacturers impose restrictions on software by scanning their barcode when changing reagents to prevent other compatible reagents.
Open system: The manufacturers do not impose restrictions on software so that reagents can be used no matter what brands.