HIV 1/2 AIDS Rapid Test Kit Near Gingival Oral Fluid For Human Immunodeficiency Virus
||2~30 º C
||Near Gingival Oral Fluid
||Rapid Test + Extraction Tube with Buffer + Swab
||Colloidal gold rapid tests
The HIV 1/2 Rapid Test is a rapid visual immunoassay for the qualitative, presumptive detection of antibodies to HIV-1/HIV-2 in human near-gingival oral fluidspecimens. This kit is intended for use as an aid in the diagnosis of HIV infection.
HIV is the etiologic agent of Acquired Immune Deficiency Syndrome (AIDS). The virion is surrounded by a lipid envelope that is derived from host cell membrane. Several viral glycoproteins are on the envelope. Each virus contains two copies of positive-sense genomic RNAs. HIV-1 has been isolated from patients with AIDS and AIDS-related complex, and from healthy people with high potential risk for developing AIDS. HIV-2 has been isolated from West African AIDS patients and from seropositive asymptomatic individuals.
The HIV 1/2 Rapid Test detects antibodies to HIV-1/HIV-2 through visual interpretation of color development on the internal strip. Recombinant HIV antigens are immobilized on the test region of the membrane. During testing, the specimen reacts with HIV antigen conjugated to colored particles and precoated onto the sample pad of the test. The mixture then migrates through the membrane by capillary action and interacts with reagents on the membrane. If there are sufficient HIV-1/HIV-2 antibodies in the specimen, a colored band will form at the test region of the membrane. The presence of this colored band indicates a positive result, while its absence indicates a negative result. The appearance of a colored band at the control region serves as a procedural control, indicating that the proper volume of specimen has been added and membrane wicking has occurred.
WARNINGS AND PRECAUTIONS
Immunoassay for in vitro diagnostic use only.
Do not use after expiration date.
The test should remain in the sealed pouch until use.
The used test should be discarded according to local regulations.
STORAGE AND STABILITY
The kit should be stored at 2-30°C until the expiry date printed on the sealed pouch.
The test must remain in the sealed pouch until use.
Keep away from direct sunlight, moisture and heat.
Do not freeze.
Care should be taken to protect the components of the kit from contamination. Do not use if there is evidence of microbial contamination or precipitation. Biological contamination of dispensing equipment, containers or reagents can lead to false results.