|Specimen:||Serum/Plasma||Usage:||In Vitro Diagnostics Use|
|Format:||Cassette||Shelf Time:||24 Months|
|Result:||Read In 15 Mins||Sensitivity:||97.9%|
Hepatitis B Hbsag rapid test kit,
Serum Plasma Hbsag rapid test kit,
Cassette hepatitis b rapid test kit
HBsAg & HBeAg Hepatitis B Antigen Rapid Detection Kit
【HBsAg & HBeAg INTRODUCTION】
HBsAg (Hepatitis B surface antigen) - A "positive" or "reactive" HBsAg test result means that the person is infected with hepatitis B. This test can detect the actual presence of the hepatitis B virus (called the “surface antigen”) in your blood. If a person tests “positive,” then further testing is needed to determine if this is a new “acute” infection or a “chronic” hepatitis B infection. A positive HBsAg test result means that you are infected and can spread the hepatitis B virus to others through your blood.
HBeAg stands for hepatitis B e-antigen. This antigen is a protein from the hepatitis B virus that circulates in infected blood when the virus is actively reproducing. A positive test for the hepatitis B e-antigen (HBeAg) means that there is an active infection with the hepatitis B virus, likely in someone with chronic hepatitis B. The virus is actively multiplying.
【RAPID TEST INTENDED USE】
HBsAg & HBeAg Rapid Test is a chromatographic immunoassay for the qualitative detection of hepatitis B Antigen in human serum/plasma. If the test is positive, this test may be done along with other tests, including the HBV DNA concentration (or viral load) test and the alanine aminotransferase (ALT) test. The ALT focuses on a specific enzyme to assess for liver function.
【RAPID TEST ADVANTAGES】
Results available in only 10 minutes
Easy to use, no need for extra instruments
Results can be read visually
Cost-effective test kit for the detection of Hepatitis B core antibodies
Test cassette with desiccant
【HBsAg & HBeAg TEST RESULT INTERPRETATION】
If only the C line develops, the result is negative or non-reactive.
If both the C and the T lines develop, the result is positive or reactive.
If no C line develops, the assay is invalid regardless of any color development on the T line as indicated below. Repeat the assay with a new device.
【STORAGE AND STABILITY】
The kit should be stored at 2-30°C until the expiry date is printed on the sealed pouch.
The test must remain in the sealed pouch until use.
Care should be taken to protect the components of the kit from contamination.
Do not use it if there is evidence of microbial contamination or precipitation. Biological contamination of dispensing equipment, containers, or reagents can lead to false results.