|Specimen:||Serum/Plasma/Whole Blood||Usage:||In Vitro Diagnostics Use|
|Format:||Cassette||Shelf Time:||Two Years|
|Principle:||Chromatographic Immunoassay||Disease Screened:||Dengue|
|Result:||Read In 10 Mins||Species Reactivity:||Human|
Dengue Rapid Test Cassette,
immunochromatographic Rapid Test Cassette,
Dengue Rapid Test Kit Cassette
Dengue Antibody Rapid Test Kit
Dengue is a viral infection transmitted to humans through the bite of infected mosquitoes. The primary vectors that transmit the disease are Aedes aegypti mosquitoes and, to a lesser extent, Ae. albopictus.
The virus responsible for causing dengue, is called dengue virus (DENV). There are four DENV serotypes and it is possible to be infected four times.
Severe dengue is a leading cause of serious illness and death in some Asian and Latin American countries. It requires management by medical professionals.
【RAPID TEST INTENDED USE】
This dengue rapid test is available as an immunochromatographic test cassette for the qualitative detection of Dengue NS1 antigen. This dengue rapid test is available as an antigen-capture immunochromatographic test cassette. This in vitro assay is for the qualitative detection of Dengue NS1 antigen in human serum, plasma and/or whole blood samples. Dengue NS1 antigen is a glycoprotein present in high concentrations in dengue-infected patients in the early clinical phase of the disease. The test is intended for professional use as a tool in the diagnosis of dengue virus infections.
【RAPID TEST ADVANTAGES】
Results ready in 10 minutes
Easy to use, Can be used with whole blood, serum, or plasma samples
Cost-effective test kit for the detection of Hepatitis B core antibodies
【RAPID TEST SPECIMEN】
Individually sealed foil pouches containing a test cassette with desiccant
【DENGUE TEST RESULT INTERPRETATION】
If only the C line develops, the result is negative or non-reactive.
If both the C and the T lines develop, the result is positive or reactive.
If no C line develops, the assay is invalid regardless of any color development on the T line as indicated below. Repeat the assay with a new device.
【STORAGE AND STABILITY】
The kit should be stored at 2-30°C until the expiry date is printed on the sealed pouch.
The test must remain in the sealed pouch until use.
Keep away from direct sunlight, moisture and heat. Do not freeze.
Care should be taken to protect the components of the kit from contamination. Do not use it if there is evidence of microbial contamination or precipitation. Biological contamination of dispensing equipment, containers, or reagents can lead to false results.