|Place of Origin:||Foshan, China|
|Minimum Order Quantity:||1000|
|Packaging Details:||25 tests/box, 1000 tests/carton|
|Delivery Time:||10-20 working days|
|Payment Terms:||T/T, Western Union, MoneyGram|
|Supply Ability:||600000 per week|
|Material:||Plastic||Shelf Life:||2 Years|
|Test Format:||Cassette||Specimen:||Whole Blood, Serum, Plasma|
|Detect Target:||Antigen NS1 Dengue Virus||Pack:||25 Tests/box|
Dengue Antigen Rapid Test Kit,
DEN NS1 Rapid Test Kit,
DEN NS1 Rapid Test Cassette
CE Dengue Antigen DEN NS1 Rapid Test Kit Cassette For Whole Blood Serum Plasma
The Dengue NS1 Rapid Test Cassette (Whole Blood/Serum/Plasma) is a rapid
chromatographic immunoassay for the qualitative detection of NS1 antigen of Dengue
virus in human whole blood, serum, or plasma as an aid in the diagnosis of Dengue
The Dengue NS1 Rapid Test Cassette (Whole Blood/Serum/Plasma) is a rapid test that
utilizes a combination of Dengue antibodies coated colored particles for the detection of
Dengue NS1 antigen in human whole blood, serum, or plasma.
【SPECIMEN COLLECTION AND PREPARATION】
The Dengue NS1 Rapid Test Cassette (Whole Blood/Serum/Plasma) can be performed using whole blood, serum, or plasma.
To collect Fingerstick Whole Blood specimens:
Wash the patient’s hand with soap and warm water or clean with an alcohol swab.
Allow to dry.
Materials required but not provided
【DIRECTIONS FOR USE】
Allow the test, specimen, buffer and/or controls to reach room temperature (15-30°C)
prior to testing.
1. Bring the pouch to room temperature before opening it. Remove the test cassette from
the sealed pouch and use it within 1 hour.
2. Place the cassette on a clean and level surface.
For Serum or Plasma specimen:
Hold the dropper vertically and transfer 3 drops of serum or plasma (approximately 75μL) to the specimen area, and start the timer. See illustration below.
For Venipuncture Whole Blood specimen:
Hold the dropper vertically and transfer 3 drops of whole blood (approximately 75 μL) to the specimen area, then add 1 drops of buffer (approximately 40 μL), and start the timer. See illustration below.
For Fingerstick Whole Blood specimen:
To use a capillary tube:
Fill the capillary tube and transfer approximately 75 μL of fingerstick whole blood specimen to the specimen area of test cassette, then add 1 drop of buffer (approximately 40 μL) and start the timer. See illustration below.
To use hanging drops:
Allow 3 hanging drops of fingerstick whole blood specimen
(approximately 75 μL) to fall into the specimen area of test cassette, then add 1 drop of buffer (approximately 40 μL) and start the timer. See illustration below.
3. Wait for the colored line(s) to appear. Read results at 10 minutes. Do not interpret the
result after 20 minutes.
【INTERPRETATION OF RESULTS】
(Please refer to the illustration above)
POSITIVE: * Two distinct colored lines appear. One color line should be in the control region (C) and another color line should be in the test region (T).
*NOTE: The intensity of the color in the test line region (T) will vary depending on the concentration of Dengue NS1 antigen present in the specimen. Therefore, any shade of red in the test region should be considered positive.
NEGATIVE: One color line appears in the control region (C). No apparent red or pink line appears in the test region (T).
INVALID: Control line fails to appear. Insufficient specimen volume or incorrect procedural techniques are the most likely reasons for control line failure. Review the procedure and repeat the test with a new test cassette. If the problem persists, discontinue using the test kit immediately and contact your local distributor.
DEWEI MEDICAL EQUIPMENT CO.,LTD
5th floor, No. 4 building, Shiyou Industrial Park, Jun'an , Shunde, Foshan, China 52832